ATTEN ADSCAL vs Rigol DSe were the two options left for considering . Manual, tracking, automatic cursor measurement function;. I have recently bought an ATTEN ADSCAL+ oscilloscope. During to instructions, I get the message: “Firmware update failed, error 0xFE. Buy Now Only: US$, Buy ATTEN ADSCAL Digital Oscilloscope MHz at the ADS Series User Manual Size: ( KB).
|Published (Last):||6 December 2018|
|PDF File Size:||2.46 Mb|
|ePub File Size:||7.66 Mb|
|Price:||Free* [*Free Regsitration Required]|
ATTEN ADS SERIES MANUAL Pdf Download.
Support USB flash drive storage. Horizon position at limit! Same with the software.
Page of Go. Do Not Operate in an Explosive Atmosphere. Page Save Store created mask setting.
By pressing ‘print’ button you will print only current page. Display the newly-acquired waveform.
Picture Sampling modes Peak Detect: Screen Normal set to normal adss1102cal. The Probe Check function verifies that the Probe attenuation option matches the attenuation of the probe.
Atten ADSCAL Oscilloscope Product Reviews
II Chapter 1 Accidence Return Press to return the waveform recorder main menu. You maual push a vertical menu button such as the CH 1 MENU buttonand select the Probe option that matches the attenuation factor of your probe. Guaranty card CD including EasyScope3. Sync Line Num All lines Select appropriate video sync.
Atten ADS1000 Series Manual
Chapter 2 Functions Instruction And Operation Chapter 2 Functions Instruction and Operation To use your oscilloscope effectively, you need to learn about the following oscilloscope functions: The oscilloscope has seen a trigger and is acquiring the posttrigger manuaal.
Therefore, the FFT spectrum can show valid frequency information higher than the oscilloscope bandwidth. If a product or CRT proves defective within the mahual period, ATTEN will provide repair or replacement as described in the complete warranty adz1102cal.
About StoreInfinity We source consumer electronics, electrical engineering tools and other electronic products. Source Select input signal channel. Triggering On A Video Signal 3. When you recall the setup, the oscilloscope will be in the mode from which the setup was saved. Page 60 Set trigger type of the vertical channel signal to Mode Pulse Pulse trigger.
It disconnects the input signal.
Pop-up Menu display pattern is more convenient for users using it. Contact the site with comments or questions. Picture Table Mask Setting function menu 1: Channel waveform and it’s FFT waveform display on split screen.
Click here for more detail Spec:.
This Chapter shows you how to operate following tasks: Page 59 The trigger signal comes from two vertical channels when you use alternative trigger. Record Press this button to enter the waveform record menu.
Page 33 Probe attenuation factor Status changes with the probe attenuation Picture 5.